[JN] Modern lightweight oscilloscopes suitable for tubeaudio?
tubegarden at aol.com
tubegarden at aol.com
Fri Dec 11 16:44:11 CST 2009
Gee, a new, modern, digital, solid state, 20 GHz, lightweight, two channel oscilloscope emulator is not a natural choice for examining signals in vacuum tube audio circuits?
What is this world coming to these days?
And why, oh God why, do you want to catch the latest wave?
You can get a killer old vacuum tube Tectronics for peanuts :) The shipping may be elephantine. You could put a sail on the cart and ride it home...
Just my two cents ;)
From: Guido Tent _ TentLabs <Guido at tentlabs.com>
To: Hans Hedal <hans.hedal at post.cybercity.dk>
Cc: joelist List <sound at lists.io.com>
Sent: Fri, Dec 11, 2009 3:37 pm
Subject: Re: [JN] Modern lightweight oscilloscopes suitable for tubeaudio?
These scopes are all digital, lack input resolution. The difference between a good analogue scope and modern lightweights is HUGE
so, if you want a good look on your signal, avoid the digital ones
at my lab, among 8 analogue scopes, we have one Fluke combiscope, for the sole reason it allows you to store pictures (and recently we aquired an AP2, dual domain
On Dec 11, 2009, at 10:22 PM, Hans Hedal wrote:
> Hi all
> Could anybody recommend one or more modern (hopefully low cost) > lightweight 2 channel benchtop oscilloscopes with a sufficient > maximum input voltage for tubeaudio?
> It/they shall be able to display voltages in both input, line, > driver and output stages.
> I have looked around in some different test equipment catalogs on > the Internet.
> It seems that normally the maximum input AC+DC voltage of a modern > (lightweight) 2 channel benchtop oscilloscope is limited to 400V > peak. I expect this is with the standard 1:1/10:1 input probe in the > maximum 10:1 position.
> If you for instance is measuring on a grounded cathode driver stage > with a high DC anode operating point, the maximum AC+DC voltage > could easily exceed 400Vp.
> It also seems that the normal maximum input sensitivity is 5V/div. > With the standard 10:1 (and 1:1) input probe and 2 x 4 = 8 screen > divisions this gives a maximum on-screen swing of 10 x 8 x 5V = > 400Vpp.
> If you have a driver stage, where you want to examine the output > headroom for a really large voltage swing, the maximum output swing > could exceed 400Vpp, and then you cannot display the full output > voltage curve on the oscilloscope screen.
> Does this mean that modern (lightweight) 2 channel benchtop > oscilloscopes doesn't have a sufficient maximum input voltage for > tube audio circuits and I shall look for something else?
> Sound mailing list
> Sound at mailman.soundlist.org
Guido at TentLabs.com
5616 GS Eindhoven
P +31 (0)40 2130 186
F +31 (0)40 2130 187
M+31 (0)6 2466 5125
Sound mailing list
Sound at mailman.soundlist.org
-------------- next part --------------
An HTML attachment was scrubbed...
More information about the Sound